The work concerns the development of aerogel radiators for RICH detectors. Aerogel tiles with a refractive index of 1.05 were tested with a RICH prototype on the electron beam on the VEPP-4M collider. It has been shown that polishing with silk tissue yields good surface quality, the amount of light loss at this surface being about 5–7%. The Cherenkov angle resolution was measured for a tile in two conditions: with a clean exit face and with a polished exit face. The number of photons detected was 13.3 and 12.7 for the clean and polished surfaces, respectively. The Cherenkov angle resolution for the polished surface is 55% worse, which can be explained with the forward scattering on the polished surface. A tile with a crack inside was also tested. The experimental data show that the Cherenkov angle resolution is the same for tracks crossing the crack area and in a crack-free area.
|Number of pages||3|
|Journal||Nuclear Instruments and Methods in Physics Research, Section A: Accelerators, Spectrometers, Detectors and Associated Equipment|
|Publication status||Published - 21 Dec 2017|
- Particle identification
- Ring imaging Cherenkov counter